|Indications for Use||The Dual Vac Pioneer Pro-Pump® is intended for the aspiration of oocytes in relation to the treatment of infertility relating to IVF and other gynecological procedures.|
The Dual Pioneer Pro-Pump® has been developed to provide a consistent pressure. The vacuum is activated by a foot pedal. The range of vacuum varies from 0-500mmHg. The Pioneer Pro-Pump® requires a reusable bottle kit (supplied).
|Recommendations for Use||For a full and detailed User Manual for our Dual Vac Pioneer Pro-Pump® please contact us at email@example.com.|
|Precautions and Warnings||Warning: To avoid risk of electrical shock, this equipment must only be connected to a supply mains
with protected earth.
Warning: Explosion Hazard – Do not use the Pioneer Pro-Pump® in the presence of a flammable anesthetic mixture with air, oxygen or nitrous oxide. Use of this instrument in such an environment may present an explosion hazard.
Caution: No modification of this equipment is allowed.
Caution: Damage to the Pioneer Pro-Pump® Motor will occur if the collection bottle overflows.
Caution: Do not operate the Pioneer Pro-Pump® if it appears to have been dropped or damaged.
Caution: Federal (U.S.A.) law restricts this device to sale, distribution, or use by or on the order of a licensed medical practitioner.
Caution: Modification of the Pioneer Pro-Pump® may compromise patient and operator safety. Refer servicing to LifeGlobal® Group personnel only.
Caution: Connect the AC Mains power cord to a properly grounded hospital-grade outlet. The Pioneer Pro-Pump® should be connected to the same elctrical circuit as other equipment in use on the patient. Outlets of the same circuit can be identified by members of the facility engineering department.